Gut Enter the current date and time with the keyboard legisativo the operation panel. Wash with clean water and see a physician immediately. Time display format Select the time display and printing format. Moves the cursor to the next character. Press Search to search files for ID. Use with other equipment: You cannot charge the battery when the power is turned on.
|Published (Last):||1 May 2005|
|PDF File Size:||17.65 Mb|
|ePub File Size:||8.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
Gut Enter the current date and time with the keyboard legisativo the operation panel. Wash with clean water and see a physician immediately. Time display format Select the time display and printing format. Moves the cursor to the next character. Press Search to search files for ID. Use with other equipment: You cannot charge the battery when the power is turned on. Make sure that the patient cable is completely dried after cleaning.
Also when the amplitude of these QRS waves is too small less than 0. Replace the SD card with a new one and retry the operation.
Insert the battery pack into the battery compartment. NOTE Do not turn the electrocardiograph power off while displaying the Data Management — Format screen and do not give an impact to the electrocardiograph. Set the recording paper as specified. The electrocardiograph is shipped from the factory with no battery pack installed.
For Nihon Kohden products marked with this symbol, contact your Nihon Kohden representative for disposal. Push the recorder cover in until it is closed firmly. If error information prints out, contact your Nihon Kohden representative. This product stores personal patient information. Only gender and age group can be entered using the number keys on the operation panel.
When a battery is inserted, the electrocardiograph automatically switches to battery power if the power cord is disconnected or there is a sudden power failure. Pull the card out straight from the SD card slot. It may cause leakage, overheating, explosion and fire. No part of this document may be reproduced, stored, or transmitted in any form or by any means electronic, mechanical, photocopied, recorded, or otherwise without the prior written permission of Nihon Kohden.
Do not use the output signal from the output connector of the electrocardiograph for a synchronization signal on a defibrillator. Connect the battery cable to the battery pack connector. Check that the electrodes are clean. Date display format Select the date display format. Select the file and press the Print ppn key. Preparation for Automatic Recording Change the following settings, if necessary.
Thermal head 30 km continuous recording When the thermal head wears out, the recording density gets weaker due to missing dots. Check the electrode attachment. Do not force the levislativo cable into the battery connector. A dotted line is continuously printed while this key is pressed. Close the recorder cover by pressing the center of the cover. Remove the cause and wait until this message disappears.
Select Hospital with the or function key. When more than one electrical instrument is used, there may be electrical potential difference between the instruments. Oegislativo Not enough free memory Overheating. Before Operation 1 Check that the instrument is in perfect operating order. Off Battery charge lamp: Related Articles
d.leg. 1150 - Regimen Disciplinario de La Pnp
In this case, temporarily changed settings are lost. Copyright Notice The entire contents of this manual are copyrighted by Nihon Kohden. Make sure that devreto electrode is completely dried after cleaning. Failure to follow this instruction causes overheating and damage to the electrocardiograph. Manual Cardiofax Press the Cancel function key to cancel.
DECRETO LEGISLATIVO 1150 PNP PDF